Common Fault Handling During Drying Of Air Sterilizer
During the drying process, the second evacuation is not performed after the first evacuation is completed.
① The drying time is less than 8 minutes.
② The negative pressure switch does not work (poor contact, no overlap). Adjust the negative pressure switch to meet the program requirements. If it still cannot be reached, replace the negative pressure switch.