The greater the capacity of the autoclave, the wider the range of use
Autoclave is the use of pressure saturated steam to quickly and reliable disinfection and sterilization equipment for products, suitable for medical and health services, scientific research, agriculture and other units, for medical equipment, dressings, glassware, solution medium, etc. for disinfection and sterilization, is an ideal equipment.
The sterilization pot uses pressurized saturated steam to sterilize medical equipment, dressings, liquid medicine, glassware, etc. It has the characteristics of excellent effect, simple operation, safety and reliability, and convenient carrying. It is suitable for general small medical treatment and laboratory units.
The items to be disinfected are properly bandaged, sequentially, with a gap between them, and placed on the sieve plate in the disinfection barrel. It is conducive to the penetration of steam and improves the sterilization effect. Add six liters of water to the main body of the sterilization pot. For continuous use, it must be after each sterilization. To make up the above amount of water to avoid a major accident caused by dry heat, put the disinfection barrel into the main body. After that, insert the hose on the cover of the autoclave into the convex tube in the disinfection barrel, correct the bolt groove of the cover and the main body, and sequentially and uniformly tighten the other wing nut, make the cover close to the main body.
The sterilization pan is placed flat on the heat source. At first, you must push the bleed valve to the vertical position to make the air inside the device escape. When there is a relatively urgent steam spray in the device. The pick should be pulled to a horizontal position. The pressure generated by the rising heat in the sterilization pot can be displayed on the pressure gauge. When the pressure in the sterilization pot reaches the required range, the heat source should be appropriately reduced immediately. Make it maintain constant pressure. At the same time, start to control the sterilization time according to different items and packaging.
Autoclave for medical equipment, dressing utensils and other tiles after the need to dry quickly, can immediately open the air release valve, so that the pressure vapor in the device quickly discharged, after the pressure gauge pointer returns to the zero position, wait for 1-2 minutes, then open the cover and continue to heat the device for 10-15 minutes. The autoclave has good sealing performance and effectively prevents the leakage of steam; The equipped pressure gauge can intuitively indicate the working pressure and corresponding temperature, which is conducive to mastering the sterilization effect. The product is equipped with a safety valve, to ensure safe use.