List of high pressure sterilizer components and their functions
The basic components of medical high pressure sterilizer. The chamber consists of an inner cavity and a coat. In hospitals and laboratories, the medical autoclaved chamber is also filled with steam, which means that by reducing condensate in the internal chamber, it reduces the burden on the internal chamber. This also reduces the time taken to complete the sterilization cycle. Just like an ordinary microwave oven or oven, all autoclaves have their own control panel. Of course, machines as complex as autoclaves do not have the same controller interface as household devices. Basically, the sterilization cycle follows a certain method provided by the autoclaved software, which starts a process consisting of multiple stages.
Thermostatic trap. The trap in the medical high-pressure sterilizer is a device used to allow water and air to be discharged from the chamber. It has always been an important part of the medical professional design of the high pressure sterilizer. Safety valve. Because the high-pressure sterilizer operates under high pressure, it is equipped with various safety measures and firm structures. Safety valve is basically a kind of fault safety device, which can protect users from danger if all other electronic programs cannot operate normally. Due to its important role in safety, it is always necessary to check and test the safety valve in advance. Cooling system. Before the waste water generated by the high pressure kettle enters the drain pipe, cool it first to avoid the damage caused by heat. Vacuum system. Vacuum system only exists in some types of autoclaves, which are used to replace all indoor air with steam. Steam generator. Steam generator is also called boiler, whose purpose is to provide steam source for the high pressure kettle without central steam source.