Application method of automatic Tabletop Autoclaves
Automatic Tabletop Autoclaves is suitable for medical and health enterprises, scientific research, agriculture and other units. It is an ideal equipment for disinfection and sterilization of medical devices, dressings, glassware, solution culture medium, etc. The Tabletop Autoclaves used in automatic desktop sterilizer is generally treated with 101.33kPa for 15~20min, even higher. The specific parameters depend on the model of the desktop sterilizer (Please consult our relevant staff). Can achieve the purpose of sterilization of articles. All high temperature resistant and humid articles, such as common culture medium, normal saline, clothes, gauze, Glass equipment, etc., can be sterilized by this method.
(1) add appropriate amount of water into the outer pot, put the articles to be sterilized into the inner pot, cover the lid of the pot and tighten the spiral symmetrically.
(2) heating makes the boiler generate steam. When the pointer of the pressure gauge reaches 33.78kPa, open the exhaust valve to discharge the cold air. At this time, the pointer of the pressure gauge drops down. When the pointer drops to zero, close the exhaust valve.
(3) continue heating, the steam in the pot increases, and the pointer of the pressure gauge rises again. When the pressure in the pot increases to the required pressure, the firepower will be reduced. According to the characteristics of the sterilized articles, make the steam pressure maintain the required pressure for a certain period of time, then cut off the power or fire of the sterilizer, let it cool naturally, then slowly open the exhaust valve to remove the residual gas, and then open the lid to take things.